Can TT-14500 Pro Micricentrifuge be Operated in A Cooling Room?
Q: Can TT-14500 PRO Microcentrifuge be operated in a cooling room?
A1: The microcentrifuge can be operated in ambient conditions of 5 – 35 °C. The permitted relative humidity must not exceed 75 %, non-condensing environment. When transferring the centrifuge from a colder into a warmer environment, the centrifuge must either be allowed to run warm for a half an hour in the cooling room or it must be allowed to warm up in the warmer environment for at least 3 hours. During this time, it must not be connected to a power source. In order to avoid damage from condensation, centrifuges should generally not be submitted to frequent temperature changes.